- Letter to the Editor
- Open Access
Changes in endotracheal tube cuff pressure in mechanically ventilated adult patients
Journal of Intensive Care volume 2, Article number: 7 (2014)
During mechanical ventilation, endotracheal tube cuff pressure should be maintained within proper range. We investigated the effect of frequent adjustment on cuff pressure in 27 mechanically ventilated patients. Cuff pressure was recorded every 2 h and was adjusted to 24 cmH2O each time. We found that cuff pressure was decreased by 4.9 ± 2.9 cmH2O from the target value. Cuff pressure decreased to less than 20 cmH2O in 45% of measurement occasions 2 h after adjusting it to 24 cmH2O.
To prevent gas leakage and aspiration, an endotracheal tube (ETT) with a cuff is generally used for mechanically ventilated patients. Because excessive cuff pressure decreases tracheal capillary perfusion, and insufficient cuff pressure leads to aspiration of oropharyngeal contents, [1–3] cuff pressure should be maintained within the proper range. Cuff pressure measurements are routinely taken every 8 to 24 h, and during the interval, air inside the cuff may escape from the ETT cuff surface or through the pilot balloon valve. It remains unknown whether, through frequent adjustment, cuff pressure can be maintained within the target range. We prospectively collected 1,846 data points of cuff pressure from 27 adult patients receiving mechanical ventilation for longer than 48 h with cuffed ETTs, standard high-volume low-pressure cuff (Hi-Lo Mallinckrodt Medical, Dublin, Ireland). Nurses measured the cuff pressure every 2 h using a cuff inflator (Cuff Pressure Gauge, VBM Medizintechnik GmbH, Baden-Württemberg, Germany) each time readjusting the pressure to 24 cmH2O. Ventilatory settings and body position remained unchanged during the 2-h interval. The study was approved by the hospital research board, which waived the requirement for informed consent for this observational study that was part of routine care.
Cuff pressure deviation from the target value (24 cmH2O) was −4.9 ± 2.9 cmH2O. Cuff pressure was below 20 cmH2O in 45% of the measurements, below 24 cmH2O in 93%, and over 30 cmH2O in 0.05% (Figure 1). Loss of cuff pressure is known to increase the risk of complications cuff pressure below 20 cmH2O is associated with the development of ventilator-associated pneumonia [1, 2, 4, 5]. Nseir et al. , measuring cuff pressure every 8 h, found that cuff pressure was maintained within recommended range (20 to 30 cmH2O) in only 18% of patients, that it was lower than 20 cmH2O at least once for 54% of patients, and that it was over 30 cmH2O at least once for 73% of patients. In our study, the cuff tended to deflate, probably owing to different patient characteristics than in Nseir’s study.
Because frequent readjustment did not prevent cuff pressure loss, we suspected that the measurement procedure itself might contribute to changes in cuff pressure. The air compressed in the cuff might escape to the measurement system during the connection procedure. Further study is needed to clarify the effect of measurement procedures in a broader variety of situations. Our study has several limitations: small population, varied observation time, and lack of evaluation of clinically significant outcomes. In conclusion, cuff pressure decreased to less than 20 cmH2O in 45% of measurement occasions taken from critically ill patients 2 h after adjusting it to 24 cmH2O.
Sengupta P, Sessler DI, Maglinger P, Wells S, Vogt A, Durrani J, Wadhwa A: Endotracheal tube cuff pressure in three hospitals, and the volume required to produce an appropriate cuff pressure. BMC Anesthesiol 2004, 4: 8. 10.1186/1471-2253-4-8
Sole ML, Su X, Talbert S, Penoyer DA, Kalita S, Jimenez E, Ludy JE, Bennett M: Evaluation of an intervention to maintain endotracheal tube cuff pressure within therapeutic range. Am J Crit Care 2011, 20: 109-117. 10.4037/ajcc2011661
Nseir S, Brisson H, Marquette CH, Chaud P, Di Pompeo C, Diarra M, Durocher A: Variations in endotracheal cuff pressure in intubated critically ill patients: prevalence and risk factors. Eur J Anaesthesiol 2009, 26: 229-234. 10.1097/EJA.0b013e3283222b6e
Pneumatikos IA, Dragoumanis CK, Bouros DE: Ventilator-associated pneumonia or endotracheal tube-associated pneumonia? An approach to the pathogenesis and preventive strategies emphasizing the importance of endotracheal tube. Anesthesiology 2009, 110: 673-680. 10.1097/ALN.0b013e31819868e0
Nseir S, Zerimech F, Fournier C, Lubret R, Ramon P, Durocher A, Balduyck M: Continuous control of tracheal cuff pressure and microaspiration of gastric contents in critically ill patients. Am J Respir Crit Care Med 2011, 184: 1041-1047. 10.1164/rccm.201104-0630OC
The authors declare that they have no competing interests.
AM, SA, HK, and YM established the design of the study, collected the data, and drafted the manuscript. TM participated in the design of the study and revised the manuscript. HI performed the statistical analysis and revised the manuscript. MN revised the manuscript. All authors read and approved the final manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
About this article
Cite this article
Motoyama, A., Asai, S., Konami, H. et al. Changes in endotracheal tube cuff pressure in mechanically ventilated adult patients. j intensive care 2, 7 (2014) doi:10.1186/2052-0492-2-7
- Cuff pressure
- Endotracheal tube
- Critically ill patients
- Mechanical ventilation