- Open Access
Temperature of gas delivered from ventilators
© Chikata et al.; licensee BioMed Central Ltd. 2013
- Received: 22 July 2013
- Accepted: 2 October 2013
- Published: 8 November 2013
The Erratum to this article has been published in Journal of Intensive Care 2013 1:16
Although heated humidifiers (HHs) are the most efficient humidifying device for mechanical ventilation, some HHs do not provide sufficient humidification when the inlet temperature to the water chamber is high. Because portable and home-care ventilators use turbines, blowers, pistons, or compressors to inhale in ambient air, they may have higher gas temperature than ventilators with piping systems. We carried out a bench study to investigate the temperature of gas delivered from portable and home-care ventilators, including the effects of distance from ventilator outlet, fraction of inspiratory oxygen (FIO2), and minute volume (MV).
We evaluated five ventilators equipped with turbine, blower, piston, or compressor system. Ambient air temperature was adjusted to 24°C ± 0.5°C, and ventilation was set at FIO2 0.21, 0.6, and 1.0, at MV 5 and 10 L/min. We analyzed gas temperature at 0, 40, 80, and 120 cm from ventilator outlet and altered ventilator settings.
While temperature varied according to ventilators, the outlet gas temperature of ventilators became stable after, at the most, 5 h. Gas temperature was 34.3°C ± 3.9°C at the ventilator outlet, 29.5°C ± 2.2°C after 40 cm, 25.4°C ± 1.2°C after 80 cm and 25.1°C ± 1.2°C after 120 cm (P < 0.01). FIO2 and MV did not affect gas temperature.
Gas delivered from portable and home-care ventilator was not too hot to induce heated humidifier malfunctioning. Gas soon declined when passing through the limb.
- Minute Volume
- Ventilator Circuit
- Heated Humidifier
- Bench Study
- Portable Ventilator
During unaided spontaneous breathing (SB), inspiratory gases are usually heated and humidified in the nasal cavity and pharynx. By the time the second bronchial bifurcation is reached, inspired gas temperature is 37°C, and absolute humidity is 44 mg/L . When patients are mechanically ventilated, however, the anatomy that provides this natural conditioning is bypassed, and it is necessary to warm and humidify medical gases. Heated humidifiers (HHs) and heat-and-moisture exchangers (HMEs) are commonly used humidifying devices. While HMEs are easy to use and have growing popularity, HHs are indicated for patients with air leakage or acute respiratory distress syndrome, especially with hypercapnia [2–4].
Sophisticated HHs monitor temperature at the water-chamber outlet and at the end of the inspiratory circuit. Feedback circuits control electric power to the water-chamber heating plate and the heating wire inside the inspiratory circuit, thus maintaining the temperature and humidity of inspiratory gas at 37°C and ensuring 100% relative humidity at the proximal end of the endotracheal tube. This control system may have untoward effects, for example, gas delivered from the ventilator is too hot, the higher temperature detected at the water-chamber outlet causes the HH to decrease electric power to the heating plate resulting in less vapors in the inspiratory gas .
As reported by Austin et al., the gas temperature of ventilator-outlet, warmed by heat generated by the ventilator, may rise as high as 46.7°C . The hottest ventilator was an ultracompact portable model, which incorporated a turbine to inhale ambient air. Expecting to see considerable improvement in power efficiency during the decade and more since their study, and a more recent model of the design, we carried out a bench study to investigate ventilator-outlet temperature, and the effects of distance from ventilator-outlet to HH and of fraction of inspiratory oxygen (FIO2) and minute volume (MV) on temperature.
Type of supply system
Type of flow sensor
PB840 + compressor
Hot wire flow sensor
(190°C ~ 200°C)
Hot wire flow sensor
(190°C ~ 200°C)
All four thermistors (NT-1000, San-You Technology Ltd, Saitama, Japan) in the ventilator circuit were two-point calibrated using a cooler/heater water source (HHC-51, Senko Medical, Tokyo, Japan). As a preliminary, each ventilator was left to run until the gas temperature of ventilator outlet became constant. We ascertained that this was achieved by all the tested ventilators by about 5 h. After confirming it, we recorded gas temperature at the four sites for 1 min. After recording the gas temperature following each change of ventilator setting, we waited for at least 30 min before collecting new data. All signals from the thermistors were processed through an analog/digital converter and saved on a computer at the sampling rate of 50 Hz/channel using data acquisition software (WinDaq, Dataq Instruments, Akron, OH, USA). Analysis of variance was performed using repeated measures ANOVA. All statistical testes were two-sided and a P value<0.01 was considered statistically significant. All statistical analysis was performed using commercial software (SPSS 11.01, SPSS, Chicago, IL, USA). Results were expressed as mean ± SD.
In this study evaluating the delivered gas temperature of various ventilators, we found variations as much as 10°C in the outlet gas temperature of different ventilators. Outlet gas temperature increased gradually after the ventilators began operating, and it took one of the ventilators as long as 5 h to provide gas at a stable temperature. Gas temperature decreased rapidly as the gas traveled up to 80 cm from the ventilator outlet. Neither FIO2 nor MV had any apparent influence on gas temperature.
Our results show that the gas temperature delivered from the ventilators was not so high as to cause this type of HH malfunctioning .
The gas temperature from ventilator is dependent on both ambient temperature and absorption of heat generated by the friction between the gas and internal pneumatic circuits and electrical circuit during the working of the ventilator. Portable ventilators usually inhale ambient air and control FIO2, so high ambient air temperature would result in correspondingly high gas temperature. Heat release may also influence gas temperature, and it could be poor with the ultracompact ventilator. All the ventilators tested performed adequately in our tests, which were carried out in an air-conditioned room with ambient temperature at 24.0°C ± 0.5°C. In an ambient temperature of 22.9°C ± 0.2°C, Lellouche et al. found gas temperature was 29.8°C ± 1.0°C, which did not lead to HH malfunction . When the ambient temperature was higher (29.0 ± 0.2°C), however gas temperature increased to 35.8°C ± 1.5°C .
Since the temperature of the working ventilator will always be greater than the ambient temperature and, in addition, heat release is more efficient when ambient temperature decreases, it remains unclear which factor influences gas temperature the most in different circumstances. We found the highest gas temperature with the PB840 supplied with the compressor, which was not surprising because the input gas temperature was greater than the ambient air temperature. Gas temperature was also relatively high with the turbine-equipped, ultracompact LTV1200. When we changed FIO2, it had no discernible influence on the medical gas temperature. Only when the temperature of oxygen gas and ambient air differs greatly, would the delivered gas temperature be affected by FIO2. This type of effect is illustrated by the delivered gas temperature from the PB840 supplied by compressor.
We found the temperature difference between MV of 5 L/min and 10 L/min was less than 5%, but we must also point out that MV was changed only by increasing the respiratory rate, from 10 to 20 breaths/min. Gas flow per se was not altered to modulate MV.
Gas travel distance from the ventilator outlet greatly influenced gas temperature. In a limb not warmed by a heating wire, lower-temperature air surrounding the tube easily carries away heat. Chikata et al. reported that the temperature of gas passing through a tube as short as 10 cm, if a heating wire is not used, can be greatly affected by ambient temperature . In the present study, gas temperature decreased as the gas traveled up to 80 cm from the ventilator outlet. How much it decreases is dependent on gas flow and the ambient air. Minute ventilation of our settings was 5 and 10 L/min, and gas temperature did not differ between them. Gas flow might hardly influence gas temperature. Ambient temperature was the most important factor, and we should be careful when we use it under high ambient temperature, such as radiology lab. HME is suitable under transporting or under high ambient temperature.
We investigated only one ventilator of each brand. As Lellouche et al. investigated three different days and obtained different results, gas temperature was not always same. We did not repeat our measurement in the present study, and we cannot deny the possibility of this issue.
In conclusion, in none of the ventilators gas temperature was high enough to cause HH malfunction. Even when gas temperature was relatively high at the ventilator outlet, it soon decreased to close to ambient temperature after traveling just 80 cm.
- McFadden ER, Pichurko BM, Bowman HF, Ingenito E, Burns S, Dowling N, Solway J: Thermal mapping of the airways in humans. J Appl Physiol 1985, 58: 564-570. 10.1063/1.335663View ArticlePubMedGoogle Scholar
- Branson R: Humidification for patients with artificial airways. Respir Care 1999, 44: 630-641.Google Scholar
- Martin C, Papazian L, Perrin G, Bantz P, Gouin F: Performance evaluation of three vaporizing humidifiers and two heat and moisture exchangers in patients with minute ventilation > 10L/min. Chest 1992, 102: 1347-1350. 10.1378/chest.102.5.1347View ArticlePubMedGoogle Scholar
- Martin C, Papazian L, Perrin G, Saux P, Gouin F: Preservation of humidity and heat of respiratory gases in patients with a minute ventilation greater than 10L/min. Crit Care Med 1994, 22: 1871-1876.View ArticlePubMedGoogle Scholar
- Lellouche F, Taillé S, Maggiore SM, Qader S, L’her E, Deye N, Brochard L: Influence of ambient and ventilator outlet temperatures on performance of heated-wire humidifiers. Am J Respir Crit Care Med 2004, 170: 1073-1079. 10.1164/rccm.200309-1245OCView ArticlePubMedGoogle Scholar
- Austin PN, Johannigman JA, Simmons LW, Camden MP, Campbell RS, Branson RD: Surface temperature of two portable ventilators during simulated use under clinical conditions. Mil Med 2001, 166: 843-847.PubMedGoogle Scholar
- Chikata Y, Imanaka H, Onishi Y, Ueta M, Nishimura M: Humidification during high-frequency oscillation ventilation is affected by ventilator circuit and ventilator setting. Pediatr Anesth 2009, 19: 779-783. 10.1111/j.1460-9592.2009.03068.xView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.