Changes in endotracheal tube cuff pressure in mechanically ventilated adult patients
© Motoyama et al.; licensee BioMed Central Ltd. 2014
Received: 8 August 2013
Accepted: 16 January 2014
Published: 31 January 2014
During mechanical ventilation, endotracheal tube cuff pressure should be maintained within proper range. We investigated the effect of frequent adjustment on cuff pressure in 27 mechanically ventilated patients. Cuff pressure was recorded every 2 h and was adjusted to 24 cmH2O each time. We found that cuff pressure was decreased by 4.9 ± 2.9 cmH2O from the target value. Cuff pressure decreased to less than 20 cmH2O in 45% of measurement occasions 2 h after adjusting it to 24 cmH2O.
KeywordsCuff pressure Endotracheal tube Critically ill patients Mechanical ventilation
To prevent gas leakage and aspiration, an endotracheal tube (ETT) with a cuff is generally used for mechanically ventilated patients. Because excessive cuff pressure decreases tracheal capillary perfusion, and insufficient cuff pressure leads to aspiration of oropharyngeal contents, [1–3] cuff pressure should be maintained within the proper range. Cuff pressure measurements are routinely taken every 8 to 24 h, and during the interval, air inside the cuff may escape from the ETT cuff surface or through the pilot balloon valve. It remains unknown whether, through frequent adjustment, cuff pressure can be maintained within the target range. We prospectively collected 1,846 data points of cuff pressure from 27 adult patients receiving mechanical ventilation for longer than 48 h with cuffed ETTs, standard high-volume low-pressure cuff (Hi-Lo Mallinckrodt Medical, Dublin, Ireland). Nurses measured the cuff pressure every 2 h using a cuff inflator (Cuff Pressure Gauge, VBM Medizintechnik GmbH, Baden-Württemberg, Germany) each time readjusting the pressure to 24 cmH2O. Ventilatory settings and body position remained unchanged during the 2-h interval. The study was approved by the hospital research board, which waived the requirement for informed consent for this observational study that was part of routine care.
Because frequent readjustment did not prevent cuff pressure loss, we suspected that the measurement procedure itself might contribute to changes in cuff pressure. The air compressed in the cuff might escape to the measurement system during the connection procedure. Further study is needed to clarify the effect of measurement procedures in a broader variety of situations. Our study has several limitations: small population, varied observation time, and lack of evaluation of clinically significant outcomes. In conclusion, cuff pressure decreased to less than 20 cmH2O in 45% of measurement occasions taken from critically ill patients 2 h after adjusting it to 24 cmH2O.
- Sengupta P, Sessler DI, Maglinger P, Wells S, Vogt A, Durrani J, Wadhwa A: Endotracheal tube cuff pressure in three hospitals, and the volume required to produce an appropriate cuff pressure. BMC Anesthesiol 2004, 4: 8. 10.1186/1471-2253-4-8View ArticlePubMedPubMed CentralGoogle Scholar
- Sole ML, Su X, Talbert S, Penoyer DA, Kalita S, Jimenez E, Ludy JE, Bennett M: Evaluation of an intervention to maintain endotracheal tube cuff pressure within therapeutic range. Am J Crit Care 2011, 20: 109-117. 10.4037/ajcc2011661View ArticlePubMedPubMed CentralGoogle Scholar
- Nseir S, Brisson H, Marquette CH, Chaud P, Di Pompeo C, Diarra M, Durocher A: Variations in endotracheal cuff pressure in intubated critically ill patients: prevalence and risk factors. Eur J Anaesthesiol 2009, 26: 229-234. 10.1097/EJA.0b013e3283222b6eView ArticlePubMedGoogle Scholar
- Pneumatikos IA, Dragoumanis CK, Bouros DE: Ventilator-associated pneumonia or endotracheal tube-associated pneumonia? An approach to the pathogenesis and preventive strategies emphasizing the importance of endotracheal tube. Anesthesiology 2009, 110: 673-680. 10.1097/ALN.0b013e31819868e0View ArticlePubMedGoogle Scholar
- Nseir S, Zerimech F, Fournier C, Lubret R, Ramon P, Durocher A, Balduyck M: Continuous control of tracheal cuff pressure and microaspiration of gastric contents in critically ill patients. Am J Respir Crit Care Med 2011, 184: 1041-1047. 10.1164/rccm.201104-0630OCView ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited.